TCSPC and Time Tagging Electronics
TCSPC and MCS board with PCIe interface
- One or two independent input channels and common synch channel (up to 84 MHz)
- Two models with either 25 ps (PICO model) or 1 ns (NANO model) base resolution
- "Long range mode" option for PICO model with 2.5 ns base resolution
- Ultra short dead time (<25 ns for PICO model, <1 ns for NANO model)
- Time tagging with sustained count rates up to 40 Mcps
- 32768 histogram channels
- Adjustable delay on each channel with 25 ps (PICO model) or 1 ns (NANO model) resolution
- Multi-stop capability for efficiency at slow repetition rates
- Programmable trigger output
- External synchronization signals for (fluorescence lifetime) imaging or other control events
- Drivers and demo code for custom programming
The TimeHarp 260 is a compact, easy to use, Time-Correlated Single Photon Counting (TCSPC) and Multi-Channel Scaling (MCS) board for the PCIe interface. It is based in a custom TDC design that offers an ultrashort dead time even at high temporal resolutions. The board is available in two versions with either 25 ps (PICO module) or 1 ns (NANO module) base resolution. The high quality and reliability of the TimeHarp 260 is expressed by a unique 5-year limited warranty.
Multiple input channels for highly flexible use
Each version of the TimeHarp 260 is available in different configurations with either one or two independent detection channels and an additional common sync input. All of them, including the sync input, can be used as independent timing channels for coincidence correlation experiments. Alternatively the common sync input can be used for TCSPC with fast excitation sources. The other input(s) can then be used as independent detector channels for TCSPC. In this case the TimeHarp 260 allows forward start-stop operation at the full repetition rate of mode locked lasers with stable repetition rate up to 100 MHz.
PICO module for high resolution TCSPC
The TimeHarp 260 PICO is designed for high timing resolution. With a digital resolution of 25 ps and a timing jitter < 20 ps it is well matched to the timing resolution of the majority of common photon detectors. All input channels are equipped with software-adjustable Constant Fraction Discriminators (CFD) sensitive on the falling edge. The ultra short dead time of the TimeHarp 260 PICO of < 25 ns per channel allows very high measurement rates. Along with the extremely low differential non-linearity of the instrument, excellent data quality can be obtained.
The histogramming time range of the TimeHarp 260 PICO can be extended up to seconds with an optional "long range mode". In this mode the base resolution of the board is switched to 2.5 ns and the dead time reduces to < 2.5 ns. This permits to study dynamics from picoseconds up to seconds with just a single board.
NANO module for MCS and deadtime-free correlation
The TimeHarp 260 NANO is designed for ultimately short dead time at a moderate time resolution. Exactly like the PICO model it can be used for coincidence correlations across all inputs or for TCSPC with light source trigger connected to the sync input. Because of the short dead time and the long histogram range it is particularly suited for classical Multi Channel Scaler (MCS) applications. Software-adjustable discriminators and polarity switches allow the TimeHarp 260 NANO to be interfaced to a wide range of signal and trigger sources. The board's multi-stop capability allows efficient recording of long-lived fluorescence and luminescence decays up to the seconds time range with correspondingly slow excitation rates, yet at very high detector count rates.
Adjustable delay in each input channel
Each input channel has an internal adjustable delay with ±100 ns range at either 25 ps resolution (PICO model) or 1 ns resolution (NANO model). This unique feature eliminates the need for specially adapted cable lengths or cable delays for changes in experimental set-ups.
Time-tagged Time-resolved modes
A Time-tagged mode for recording of individual photon events with their arrival time on all channels allows the most sophisticated offline analysis of the photon dynamics. Time-Tagged Time-Resolved (TTTR) data can also be correlated in real-time for monitoring of FCS experiments at count rates up to 1.000.000 counts/sec. External marker signals can be used to synchronize the device with other hardware such as scanners, e.g., for Fluorescence Lifetime Imaging (FLIM).
Programmable trigger out
The TimeHarp 260 also features a trigger output, which can generate pulse periodes between 0.1 µs and 1.678 s, corresponding to repetition frequencies between 0.596 Hz and 10 MHz. This feature can, e.g., be used to control external lasers.
Easy-to-use software or custom programming
The TimeHarp 260 software for Windows provides functions such as the setting of measurement parameters, display of results, loading and saving of measurement parameters and measurement curves. Important measurement characteristics such as count rate, count maximum, position and peak width are displayed continuously. A comprehensive online help function shortens the users' learning curve. A library for custom programming, e.g., with LabVIEW is also available as an option. Software upgrades for extended functionality will be available with further product development.
Measurement data from the TimeHarp 260 can be analyzed by different software packages. For multi-exponential reconvolution the FluoFit software is an ideal tool. For the analysis of TTTR data (e.g., FLIM, FCS, FLCS, FRET, BIFL, etc.) the SymPhoTime 64 software suite is the tool of choice.
The TimeHarp 260 can be used in different operation modes:
- Integration mode
- Oscilloscope mode
- Time-Resolved Emission Spectra (TRES)
- Time-Tagged Time-Resolved mode (listing each event arrival time)
- On-line ("real-time") correlator for FCS
Current software and Developer's Library version: 2.0
The new software version features improvements for stability and performanceand also includes the a new future proof file format in terms of maintenance and compatibility across new versions. It also supports a new hardware feature to change the dead time of the input channels for suppression of some detector artefacts and provides an updated device driver to meet new Windows driver signing requirements..
|TimeHarp 260 PICO||TimeHarp 260 NANO|
|Input channels and sync||Constant Fraction Discriminator (CFD) on all inputs, software adjustable||constant level trigger on all inputs, software adjustable|
|Number of detector channels
(in addition to sync)
|1 or 2||1 or 2|
|Input voltage range (pulse peak into 50 Ohms)||0 to -1200 mV, optimum: -100 mV to -200 mV||-1200 mV to +1200 mV|
|Input voltage max. range (damage level)||±1500 mV||±2500 mV|
|Trigger edge||falling edge||falling or rising edge, software adjustable|
|Trigger pulse width||0.5 to 30 ns||> 0.5 ns|
|Trigger pulse required rise/fall time||2 ns max.||--|
|Time to Digital Converters|
|Minimum time bin width||25 ps
in optional "long range mode": 2.5 ns
|Electrical time resolution||< 20 ps (rms)
in optional "long range mode": <1 ns (rms)
|< 500 ps (rms)|
|Dead time||< 25 ns
in optional "long range mode": <2.5 ns
|< 1 ns|
|Sustained count rate per input channel||40 × 106 counts/sec||40 × 106 counts/sec|
|Maximum sync rate (periodic pulse train)||100 MHz||100 MHz|
|Adjustable delay range for each input channel||±100 ns, resolution 25 ps||±100 ns, resolution 1 ns|
|Differential non-linearity||< 2% peak, <0.2% rms (over full measurement range)||< 2% peak, <0.2% rms (over full measurement range)|
|Count depth||32 bit (4.294.967.296 counts)||32 bit (4.294.967.296 counts)|
|Maximum number of time bins||32768||32768|
|Full scale time range||819 ns to 1.71 s (depending on chosen resolution: 25 ps, 50 ps, 100 ps, ..., 52.42 µs)
in optional "long range mode": 81.92 µs to 171 s (depending on chosen resolution: 2.5 ns, 5 ns, 10 ns, ..., 5.242 ms)
|32.8 µs to 68.48 s (depending on chosen resolution: 1 ns, 2 ns, 4 ns,...,2.09 ms)|
|Acquisition time||1 ms to 100 hours||1 ms to 100 hours|
|Sustained throughput (sum of all channels)||typ. 30×106 events/sec (depending on host PC configuration and performance)||typ. 30×106 events/sec (depending on host PC configuration and performance)|
|T2 mode resolution||25 ps||1 ns|
|T3 mode resolution||25 ps, 50 ps, 100 ps,..., 52.42 µs
in optional „long range mode“: 2.5 ns, 5 ns, 10 ns, ..., 5.242 ms
|1 ns, 2 ns, 4 ns,..., 2.09 ms|
|FiFo buffer depth (records)||8.388.608||8.388.608|
|Acquisition time||1 ms to 100 hours||1 ms to 100 hours|
|Sustained throughput (sum of all channels)||typ. 40×106 events/sec (depending on host PC configuration and performance)||typ. 40×106 events/sec (depending on host PC configuration and performance)|
|only available along with optional long range mode||always enabled|
|Period||programmable, 0.1 µs to 1.678 s (0.596 Hz to 10 MHz)||programmable, 0.1 µs to 1.678 s (0.596 Hz to 10 MHz)|
|Pulse width (typical)||10 ns||10 ns|
|Baseline level (typical)||0 V||0 V|
|Active level (pulse peak)||-0.7 V||-0.7 V|
|External marker inputs|
|Number||4 (only available in models with 2 detection channels)||4 (only available in models with 2 detection channels)|
|Input type||TTL, < 10 ns rise/fall time, > 50ns pulse width||TTL, < 10 ns rise/fall time, > 50ns pulse width|
|PC interface||PCIe 1.1 x1||PCIe 1.1 x1|
|PC requirements||Dual core CPU, min. 1.5 GHz CPU clock, min. 1 GB memory||Dual core CPU, min. 1.5 GHz CPU clock, min. 1 GB memory|
|Operating system||Windows 7/8||Windows 7/8|
|Power consumption||< 15 W from internal power supply of the PC||< 15 W from internal power supply of the PC|
All Information given here is reliable to our best knowledge. However, no responsibility is assumed for possible inaccuracies or omissions. Specifications and external appearances are subject to change without notice.
Operation and Software Features (Current version: 2.0)
The TimeHarp 260 can be used in various operation modes:
Measurement of the accumulated events as a function of time, manually started, stopped manually or upon overflow or expiration of a chosen collection time or upon reaching of a defined number of counts. 512 curves with up to 65536 time channels, more than 4 billion counts per time channel (32 bits).
Repetitive measurement and on-line display, very useful for optical alignment. Flicker free histogram display updates and large rate meters for work in the distance.
Time-Resolved Emission Spectra (TRES)
An optional hardware and software add-on allows to control a monochromator from within the HydraHarp software, allowing automated measurement of Time-Resolved Emission Spectra. This mode drives a monochromator via a stepper motor for collection of spectrally resolved lifetime histograms. Data is collected as in standard Integration Mode and saved in different blocks of memory for each wavelength. Four different monochromator types are currently supported: Sciencetech 9030, Sciencetech 9055, Acton Research SP-2155 and Acton Research SP-275.
Continuously recording events without onboard histogramming straight to disk. Together with the channel number the arrival time of each event with respect to the beginning of the experiment is recorded for ultimate flexibility in offline data analysis, e.g., in single molecule detection and Burst Integrated Fluorescence Lifetime (BIFL) measurement as well as time-resolved FCS, coincidence correlation (antibunching, etc.) or time-interval analysis. A real-time correlator is included in the TTTR mode which can be extremely useful in setting up and monitoring of FCS experiments. Fast transfer and a large FIFO buffer allow huge count rates without any loss of data. Up to four different external synchronization signals ("markers") can be fed into the data stream and allow to synchronize the data acquisition with external hardware such as scanners for Fluorescence Lifetime Imaging (FLIM).
The TimeHarp 260 software allows the control of all measurement parameters provided by the TimeHarp 260 module. The input channels are programmable for a variety of signal types. All functions of the system are controlled by a software interface for Windows 7 and Windows 8. The software provides functions such as the setting of measurement parameters, display of measurement results, loading and saving of measurement parameters and measurement curves. Important measurement characteristics such as count rate, count maximum and position, and histogram width (FWHM) are displayed continuously. A comprehensive online help function shortens the user's learning curve. Software upgrades for extended functionality will be available with further product development.
A library (DLL) for custom program development is also provided and allows to build your own applications, e.g., in LabVIEW , C++, Delphi or Visual Basic. Demo code is provided for an easy start. A Linux version will shortly be available. The libraries are API compatible, so that applications can easily be ported between the platforms.
Time-Tagged Time-Resolved (TTTR) mode allows the recording of individual count events directly to hard disk or computer memory. The timing of each photon is captured as an event record without any early data reduction (such as on-board forming of histograms). This mode is particularly interesting where e.g. the dynamics in a fluorescence process are to be investigated in depth. The availability of the full timing information permits photon burst identification, which is of great value e.g. for single molecule spectroscopy in a liquid flow. Other typical applications are Fluorescence Correlation Spectroscopy (FCS) and Burst Integrated Fluorescence Lifetime (BIFL) measurements. Together with an appropriate scan controller, TTTR mode is also suitable for ultra fast Fluorescence Lifetime Imaging (FLIM) with unlimited image size. Applications beyond fluorescence spectroscopy are e.g. time interval analysis, quantum optics and related basic rearch. The TimeHarp 260 actually supports two different time-tagging modes, T2 and T3 Mode - a concept originally introduced with the PicoHarp 300 and HydraHarp 400. They differ slightly in their use of the input channels and by using the suitable mode, a very wide range of applications can be covered.
In T2 Mode all signal inputs of the TimeHarp 260 are functionally identical. There is no dedication of one channel to a sync signal. All inputs can be used to connect photon detectors. The events from all channels are recorded independently and treated equally. In each case an event record is generated that contains information about the channel it came from and the arrival time of the event with respect to the overall measurement start. If the time tag overflows, a special overflow marker record is inserted in the data stream, so that upon processing of the data stream a theoretically infinite time span can be recovered at full resolution. Dead times exist only within each channel but not across the channels. Therefore, cross correlations can be calculated down to zero lag time. This allows powerful new applications such as FCS with lag times from picoseconds to hours to be implemented with one instrument. Autocorrelations can also be calculated at the full resolution but of course only starting from lag times larger than the deadtime.
The T3 Mode is specifically designed to use periodic sync signals from pulsed lasers with high repetition rate up to 100 MHz. This signal is connected to the dedicated sync channel. As far as the experimental setup is concerned, this is similar to classic TCSPC in histogramming mode. In addition to the picosecond start-stop timing, the channel number is recorded and each event is time tagged with respect to the beginning of the experiment. The time tag is obtained by simply counting sync pulses. From the T3 Mode event records it is therefore possible to precisely determine which sync period a photon event belongs to. Since the sync period is also known precisely, this furthermore allows to reconstruct the arrival time of the photon with respect to the overall experiment time. If the counter overflows, a special overflow marker record is inserted in the data stream, so that upon processing of the data stream a theoretically infinite time span can be recovered.
External Event Markers
Both dual channel versions of the TimeHarp 260 support capturing up to four external marker events in the TTTR modes that can be fed to the instrument as TTL signals via a suited SubD connector. These events are recorded as part of the TTTR data stream. This allows to precisely synchronize the TTTR measurement with almost any experiment. The most important applications of this feature are FLIM and FRET imaging. This concept is used in the cutting-edge time-resolved mircroscope MicroTime 200.
The acquisition software provided with the instrument comes with a rich set of demo programs that enable users to write their own analysis and display programs for TTTR data. Users who prefer to use standard data analysis algorithms out of the box may want to consider the powerful SymPhoTime 64 software suite. It implements a wide range of state-of-the-art analysis algorithms for FLIM, FCS and FRET to name only a few.
The TimeHarp 260 can be used for various applications that can make use of a multi-channel TCSPC and/or time tagging system with independent channels, such as:
- Time-Resolved Fluorescence
- Fluorescence Lifetime Imaging (FLIM)
- Phosphorescence Lifetime Imaging (PLIM)
- Fluorescence Correlation Spectroscopy (FCS)
- Fluorescence Lifetime Correlation Spectroscopy (FLCS)
- Foerster Resonance Energy Transfer (FRET)
- Stimulated Emission Depletion Microscopy (STED)
- Dual Focus Fluorescence Correlation Spectroscopy (2fFCS)
- Pulsed Interleaved Excitation (PIE)
- Fluorescence Anisotropy (Polarization)
- Single Molecule Spectroscopy / Detection
- Singlet Oxygen
- Time-Resolved Photoluminescence (TRPL)
- TRPL Imaging
- Fluorescence Upconversion
- Bunch Purity
- Coincidence Correlation
- Quantum Communication
- Quantum Entanglement
- Quantum Teleportation
- Quantum Information Processing
- Positron Annihilation Lifetime Spectroscopy (PALS)
- Time response characterization of optoelectronic devices
- Thomas-Bollinger single photon method
- Muon lifetime experiments
The following documents are available for download:
- Datasheet TimeHarp260
- Brochure about PicoQuant's photon counting and timing products
- Reference publication: Integrated multichannel photon timing instrument with very short dead time and high throughput
- Technical note: Time-Correlated Single Photon Counting (TCSPC)
Latest 10 publications referencing TimeHarp 260, TimeHarp 100/200
The following list is an extract of 10 recent publications from our bibliography that either bear reference or are releated to this product in some way. Do you miss your publication? If yes, we will be happy to include it in our bibliography. Please send an e-mail to email@example.com containing the appropriate citation. Thank you very much in advance for your kind co-operation.